JY-1S high precision magnetic stirring oil bathProduct Categories:
Laboratory equipment / water bath
JY-1S high precision magnetic stirring oil bath
|JY-1S high-precision magnetic stirring oil bath is an improvement of our factory based on DF-1 collector-type magnetic stirrer. The machine has large stirring capacity, high temperature control accuracy, and dry burning, boiling water, and burning. Three functions of oil. The heating through the bottom has uniform heating, and the magnetic stirring function is added, which expands the use of this instrument. It is especially suitable for stirring experiments at high temperature in rotary evaporators, spherical bottles, and various glass burners. It is one of the standing instruments in modern laboratories.|
|Details of JY-1SJY-1S high precision magnetic stirring oil bath pot :|
Product overview of high-precision magnetic stirring oil bath :
JY-1S high - precision magnetic stirring oil bath is an improvement of our factory based on DF-1 collector-type magnetic stirrer. The machine has large stirring capacity, high temperature control accuracy, and dry burning, boiling water, and burning. Three functions of oil. The heating through the bottom has uniform heating, and the magnetic stirring function is added, which expands the use of this instrument. It is especially suitable for stirring experiments at high temperature in rotary evaporators, spherical bottles, and various glass burners. It is one of the standing instruments in modern laboratories.
Product Features of High Precision Magnetic Stirring Oil Bath :
、锅体由优质不锈钢一次成型压制而成，无焊缝，内部办圆角过度，发热部件、传感器安装在锅体以外，锅内极易清洗，304不锈钢材质，永不生锈。 1. The pot body is made of high-quality stainless steel at one time, without welding seams. The internal corners are excessive. The heating components and sensors are installed outside the pot body. The pot is easy to clean. The 304 stainless steel material will never rust.
、钢化玻璃盖子，内部工作情况一目了然。 2. Tempered glass cover, the inner working condition is clear at a glance.
、先进的微电脑温控仪，集定时、控温、温度显示功能于一体。 3. Advanced microcomputer temperature controller, which integrates timing, temperature control and temperature display functions.
、具有超温报警，超温保护功能。 4 , with over-temperature alarm, over-temperature protection.
、传感器测温误差跳出功能。 5 , sensor temperature error jump out function.
、双屏LED显示屏，时间、温度一目了然。 6 , dual-screen LED display, time and temperature at a glance.
、集成调速电路，搅拌速度无级可调，运转平稳。 7 , integrated speed control circuit, stepless adjustment of stirring speed, smooth operation.
Third, high-precision magnetic stirring oil bath technology instructions:
、电 1.Electricity Source: Single-phase AC 220V ± 10V 50HZ
、整机功率： 860W 2. Overall power: 860W
3. Stirring power: 40W
4. Heating power: ≤800W
、恒温设定与测量：微电脑控制，LED双屏显示 5. Constant temperature setting and measurement: microcomputer control, LED dual screen display
、调速范围：起动～2000转／分，无级调速 6. Speed adjustment range: starting ~ 2000 rpm, stepless speed adjustment
7. Temperature control range: RT-300.0 ℃. Accuracy ± 0.1 ℃
8. Temperature fluctuation: ≤0.2 ℃
、温度定时范围：1-9999分钟或常开 9 , temperature timing range: 1-9999 minutes or normally open
、工作尺寸：Φ235×140mm 10. Working size: Φ235 × 140mm
、工作容量：5升 11. Working capacity: 5 liters
Fourth, the use of high-precision magnetic stirring oil bath pot :
、在首次使用时请检查仪器外观是否完好，电源线应无破损。 1. Please check whether the appearance of the instrument is intact and the power cord should not be damaged when using it for the first time.
、将导热油加入锅内，然后接通电源。 2. Add the heat-conducting oil to the pot, and then turn on the power.
、将搅拌子放入盛杯溶液中，然后将烧杯放置在锅的中央。 3. Put the stirrer in the cup solution, then place the beaker in the center of the pot.
、打开电源开关，数字温控亮，上排显示实际温度，下排显示设定温度，点击功能键set，此时上排显示sp，下排显示设定温度，点击△或▽键设定工作温度，设定完毕再次点击功能键set，此时上排显示st，下排显示恒温定时时间，点击△或▽键选择恒温定时时间，时间设定完毕，点击功能键退出设定状态，此时该仪器将按设定温度和时间运行。 4. Turn on the power switch, the digital temperature control is on, the upper row displays the actual temperature, and the lower row displays the set temperature. Click the function key set, and the upper row displays sp, and the lower row displays the set temperature. Working temperature. After setting, click the function key set again. At this time, the upper row displays st, and the lower row displays constant temperature timing time. Click the △ or ▽ key to select the constant temperature timing time. The instrument will run at the set temperature and time.
、打开搅拌开关，将调速旋钮按顺时针方向缓慢旋转，搅拌转速由慢到快。 5. Turn on the stirring switch, and slowly rotate the speed control knob clockwise to slow the stirring speed from slow to fast. Adjust to the required speed. If the stirrer beats during work, you can adjust the speed control knob counterclockwise to the lowest speed, and then repeat the previous operation to resume normal stirring.
、使用完毕，将速度旋钮调至最低端，然后关闭搅拌开关。 6 , after use, turn the speed knob to the lowest end, and then turn off the stirring switch.
、关闭电源开关，切断电源，待油温冷却后取出烧杯即可。 7. Turn off the power switch, cut off the power, and take out the beaker after the oil temperature has cooled.
Five, high-precision magnetic stirring oil bath pot matters needing attention:
、为确保安全，请使用三孔安全插座，使用时必须妥善接地。 1. To ensure safety, please use a three-hole safety socket, and it must be properly grounded during use.
、加热时间一般不宜过长，间歇使用能延长寿命。 2. The heating time should not be too long. Intermittent use can extend the life. It can work continuously for 8 hours at medium speed and 4 hours at high speed.
3. The instrument will have different heating parameters when using different heating media. The parameters are adjusted as follows:
First press and hold the function key set for about 4 seconds. When the temperature controller displays Lc, let go of the handle. Click the △ key to adjust the parameter to 3. Press the function key set continuously to find the P parameter. Yes, after the adjustment is completed, long press the function key set, and release the hand when the actual temperature is displayed in the upper row. At this time, the internal heating parameters set will be saved.
Parameter reference value: (water: P = 15) (oil: P = 25) (dry burning: P = 75)
4. The oil level should not be too high to prevent experimental errors caused by oil spills.
5. In order to extend the service life of the instrument, please be careful not to get moisture in the control box to prevent leakage.
、长时间不用，请擦试干净置于干燥通风处。 6. If it is not used for a long time, please wipe it clean and put it in a dry and ventilated place.
- company name:
- Jintan Tianjing Experimental Instrument Factory
- Ouyang Jun
- contact address:
- No. 22 Hongshan Road, Jincheng Industrial Park, Jintan City, Jiangsu Province
- Postal code: