Oil bathProduct Categories:
Laboratory equipment / water bath
HH-W603 High-precision circulating oil bath (0.01 ℃)
|The inner shell of the high-precision circulating oil bath pot is all made of 304 stainless steel, and the surface of the outer shell is electrostatically sprayed again, which greatly improves its anti-corrosion performance. Five-digit digital tube display, dual screen display, display resolution up to 0.01 ℃.|
|HH-W603HH-W603 high precision circulating oil bath (0.01 ° C) details:|
简介： Brief introduction of high precision circulating oil bath :
In order to meet the user's increasing requirements for the accuracy and fluctuation of the experimental temperature, HH-W603 is an improvement of our factory based on HH-W602 super constant temperature oil bath . In addition to all the characteristics of HH-W602, With higher temperature control accuracy.
特点： Features of high-precision circulating oil bath :
◇ The shell of the shell is all made of 304 stainless steel, and the surface of the shell is electrostatically sprayed again, which greatly improves its anti-corrosion performance.
◇ Five-digit digital tube display, dual screen display, display resolution up to 0.01 ℃.
◇ Using ultra-fine glass fiber as insulation material, no matter how high the working temperature is, the casing has no obvious thermal sensation.
◇ Long oil pump design, the motor can work continuously under high temperature.
◇ Anti-room temperature protection system, with over-temperature sound and light alarm function when the system has an accident, and cut off the heating power.
◇ 485 interface can be configured according to user requirements, effectively connect to a computer or printer, and real-time recording can be realized (optional plus 1000 yuan).
◇ Low noise design, no-load noise is less than 35dB.
技术参数： Technical parameters of high-precision circulating oil bath :
◇ Model: HH-W603
◇ Temperature control range: room temperature -300.00 ℃
◇ Temperature control accuracy: 0.01 ℃
◇ Temperature wave degree: <0.02 ℃
◇ Temperature uniformity: 0.02 ℃
◇ Timing range: 1-99999 minutes
◇ Heating power: 2000W
◇ Circulating pump power: 40W
◇ Circulation flow: 6L / min
◇ Working size: 400 × 300 × 180mm
◇ Opening size: 300 × 265mm
◇ Slot depth: 180mm
使用方法： Use of high-precision circulating oil bath :
◇ The instrument should be placed on a flat workbench before installation, and the appearance should be checked first: the appearance should not be damaged, the appearance of the instrument should be complete, the wire insulation should be good, the plug should be intact, and the power switch should be flexible. 超级恒温油 浴使用220V交流电源，请在使用前确定电源插座额定电流不小于10A，并具有安全接地装置(请使用时工作室内接入接地装置)。 The power cord of each device is connected to a single-phase three-pole plug. The top electrode of the plug is a ground electrode. Users must use a single-phase three-pole power socket. The ground of the power socket should be reliably grounded. The super constant temperature oil bath uses 220V AC power. Before use, please make sure that the rated current of the power socket is not less than 10A, and it has a safety grounding device (please connect the grounding device in the working room when using it).
◇ Pour the oil into the oil tank, (please choose different oil, such as methyl silicone oil, phenyl silicone oil, high temperature heat conductive oil) according to the different temperature.
◇ Put in the container or items to be heated.
◇ Turn on the power and turn on the power switch. At this time, you can click the function key (set) on the thermometer, Sp appears, and then click △ or ▽ to select the temperature to be heated. If you need to adjust the temperature quickly, you can use the ← key (left (Shift key) Quickly adjust the hundred, ten, and one digits respectively. After the setting is completed, click the set key again. At this time, the word ST appears. At this time, the timing can be adjusted. The adjustment method is the same as before, and finally click the set key The above parameters will be saved and the instrument will run according to the set parameters.
◇ Turn on the circulation pump switch, at this time the circulation pump starts to work, thereby ensuring the uniformity of the oil temperature.
◇ After use, turn off the circulation pump first, and then turn off the power. The last set parameters will be saved.
使用注意事项： Precautions for using high-precision circulating oil bath :
◇ Because the instrument works at high temperature, please wear protective gloves when operating.
◇ When not in use for a long time, please purge the oil in the box and do maintenance work.
◇ The instrument must use a three-hole safety socket, and must have a safety grounding device, and the capacity must not be less than 10A.
◇ In order to ensure the accuracy of oil temperature, different heating oils must be used for different working temperatures.
The warranty period of the product is one year from the date of shipment, and it will be repaired for life.
Note: can be customized according to user requirements
- company name:
- Jintan Tianjing Experimental Instrument Factory
- Ouyang Jun
- contact address:
- No. 22 Hongshan Road, Jincheng Industrial Park, Jintan City, Jiangsu Province
- Postal code: